NHS Orders AcceptedInstant account for Gov Entities / NHS / School
Fast Free UK DeliveryFor Healthcare Facilities
30 Day Product GuaranteeOn All Purchase Orders
Qimoharpoon is a closed system transfer device that has an air-intake tube connected
Qimoharpoon is a closed system transfer device that has an air-intake tube connected to a 50ml balloon filled with sterile air which ensures pressure equalisation
Qimono the complete closed system for safe delivery of cytotoxic drugs
Stay safe with Qimono
It is widely documented that exposure to Systemic Anti-Cancer Therapy (SACT), can pose a significant health risk to patients during treatment as well as healthcare workers involved in the preparation and administration of the drugs .
Many SACT agents are known to be carcinogenic, teratogenic and mutagenic. Exposure may be through skin contact, skin absorption, inhalation of aerosols and drug particles as well as ingestion and needlestick injuries. This can occur at various points of the treatment process: during the preparation of the medication, whilst the drugs are administered and after therapy .
Adverse health effects reported by chemotherapy nurses include headaches, dizziness, nausea and hair loss.
Description of the device
QimoHarpoon is a device designed for the reconstitution and withdrawal of high-risk medications. With QimoHarpoon, it is possible to dissolve powdered drugs or lyophilisates and to repeatedly withdraw drugs with a syringe from the same vial/bottle. QimoHarpoon is a needle-free device with a metallic tube (acting as air-intake device) connected to a balloon (50 cc volume). The function of the balloon is to prevent the emission of toxic vapour from the vial and to allow injection and drug aspiration while equalizing pressure in a closed system.
QimoHarpoon is made of: _ a harpoon-shaped spike which perforates the vial/bottle stopper and which cannot be removed from the vial. An opening is located on the harpoon which facilitates injection of the fluid for reconstitution and allows the drawing-up of the final reconstituted solution. From this opening the very last drops of drug can be drawn-up by pulling delicately until the harpoon stops against the cap of the vial. _ a mobile metallic tubing, acting as an air-intake device connected to a balloon filled with sterile air. When this metallic tube is pushed to the bottom of the vial*, the pressure will be equalized and the toxic vapours confined within the balloon. _ a tight protective cover between the base of the spike and the neck of the vial to avoid any contamination with fluids which could leak from the cap of the vial. _ a Qimo Female port, needle-free device, multi-purpose, closed IV connector. Qimo Female can be accessed with any male Luer or Luer-lock syringe or with the Luer bayonet end of Qimo Male. Qimo Female: see technical file code 7210.02. Qimo Male: see technical file code 7210.91.
Code 7213.02: QimoHarpoon _ Adapted for vials with a maximum height of 9 cm. Code 7213.01: QimoHarpoon S _ Adapted for vials with a maximum height of 6 cm.